What should be done if the Control Solution received does not correlate to the values for the lot of test strips received? How is a blood sample collected? Use capillary tubes or collectors to collect fingerstick whole blood. Venous whole blood samples can be collected in EDTA tubes lavender tops or heparin tubes green tops. How much blood is needed for each test strip?
|Published (Last):||14 April 2007|
|PDF File Size:||10.56 Mb|
|ePub File Size:||8.6 Mb|
|Price:||Free* [*Free Regsitration Required]|
Test Strips The MEMo Chip Specimen Collection Page 3 When to Replace the Batteries The MEMo Chip reads the test strip expiration date, tells the analyzer which test s to run, contains the calibration curve and the lot number for the specific test strip lot, controls test sequences and timing, and establishes the measuring range for the test. Page 6: Venous Whole Blood Samples 5. Gently apply pressure to the finger near the first knuckle of the ring or middle finger and engorge the end of the finger with a squeeze and release technique.
Use the first drop of blood to immediately dose the tip of the eGLU test strip until the channel is filled and the screen shows a beating heart icon over the glucose strip. Page 7: Equipment A. Equipment 1. Materials 1. Each check strip is calibrated to read a specific reflectance.
Page Expected Results Immediately replace the vial caps, making sure the test strip vials are capped tightly. This indicates the system is ready for a sample control to be applied to the eGLU test strip first.
Page Expected Results 8. The blood drop should be a large drop at the fingerstick site to touch the tip of the capillary tube.
With 2 to 3 squeeze-and-release motions, allow the horizontal fill of the capillary tube without the introduction of bubbles or air spaces. When the blood drop icon is over the Lipid Panel test strip, use the filled capillary tube and plunger to hover over the blood application window of the test strip and deposit the entire sample onto the blood application window while hovering.
Cardiochek PA Tragbares Vollblut-Testsys
CardioChek PLUS User Manual